DRPL-II plate heat flow meter thermal conductivity tester (plate heat flow meter method)
DRPL-II Product model: DRPL-II
2019/4/24 0:00:00 Product time: 2019/4/24 0:00:00
DRPL-II导热系数测试仪采用在试样一面加入稳定的热面温度，热量通过试样传递到冷面（室温），测量传递的热流来计算导热系数和热阻。 Brief description: The DRPL-II thermal conductivity tester uses a stable hot surface temperature on the sample side, and the heat is transferred to the cold surface (room temperature) through the sample. The transferred heat flux is measured to calculate the thermal conductivity and thermal resistance.
DRPL-II thermal conductivity tester (plate heat flow meter method)
I. Overview: DRPL-II thermal conductivity tester (plate heat flow meter method) adopts the internationally popular heat flow meter to detect the thermal conductivity and thermal resistance method, with a computer to achieve automatic detection, and generate an experimental report.The instrument uses a stable hot surface temperature on one side of the sample, and the heat is transferred to the cold surface (room temperature) through the sample. The transferred heat flux is measured to calculate the thermal conductivity and thermal resistance. This test method is simple, fast, and repeatable.Very suitable for research and development of heat transfer in metal materials such as profiles. 、气凝胶板、真空玻璃、 等测试，广泛应用在大专院校、科研院所、质检、厂矿。It can also be used fortesting ofplastics, rubber, graphite, insulation materials , aerogel panels, vacuum glass,etc. It is widely used in colleges, scientific research institutes, quality inspection, factories and mines. Reference standards: GB / T10295-2008 (Determination of steady-state thermal resistance and related characteristics of thermal insulation materials, heat flow meter method) and ASTM C518-04 Test method for determining steady-state heat flux and heat transfer characteristics by heat flow meter method.
Second, the main technical parameters: 0.001～300W/mk。1. Thermal conductivity range: 0.00001 ～ 10W / mk,0.001 ～ 300W / mk. 2. Measurement accuracy: 3% (domestic heat flow meter); 1% (imported heat flow meter); 3. Sample size: 20 × 20 ～ 300 × 300 (mm).Special sizes can be customized. 4, can test solid materials, powdery materials. 5. The measurement time is 15-20min. 6, hot surface temperature and control: room temperature -199.9 ℃ (standard configuration), room temperature-499.9 ℃, higher temperatures can be determined according to user requirements.Flat heater, bidirectional thyristor control, temperature control accuracy 0.1 ℃ 7, cold surface temperature and control: 0-99.9 ℃ (standard configuration), -10-99.9 ℃, lower temperature can be determined according to user requirements.The cold surface adopts water cooling to ensure the stable temperature of the cold surface (precision constant temperature water tank, temperature adjustable), and the temperature control accuracy is 0.1 ℃. 8. The whole process uses a computer to automatically test and store and print experimental data.